IDDLL now returns DPI correctly, so measurements can be placed on an image without calibrating. (F780)
Version 10.0 Build 2
Release Date: 5/27/2010
Resolved crash issue if ActiveX control is not installed.
Renamed DDE entry.
Resolved status updating for “Click Start Acquisition” notification.
Removed icon from Start Acquisition button.
Adjusted tile rotation on sensor capture.
Resolved bug when first capturing from layout upon opening software.
Version 10.0 Build 1
Release Date: 5/5/2010
Components and environment updated for compliance with new versions of the software.
If the video window is closed, Acquisition mode is now automatically ended.
Reworded video freezing status.
When Gendex Imaging is re-opened, log is now cleared, and selected hardware is redisplayed.
Added support for GxPicture 3.0.
Version 1.0 Build 20
Release Date: 3/9/2010
Added support for GxPicture 2.8.
Added better names for IO cameras
Version 1.0 Build 19
Release Date: 12/18/09
Corrected gamma application settings between ceph/pan/io images.
Version 1.0 Build 17
Release Date: 12/5/07
Strange inversion of images corrected. Problem was due to an inverted palette being returned from the hardware and the base software expecting standard grayscale palettes.
Memory leak that occurred during certain image improvement processes corrected.
Automatically creates the “c:\\VixTemp” folder on the target computer if it does not exist.
New feature allowing the extension to return an image type and subtype to the invoking application. These types can be customized in the IDDLL.
Version 1.0 Build 15
Release Date: 1/17/07
Added 16-bit image support.
Version 1.0 Build 12
Release Date: 7/10/06
Added invert grayscale images option to IntraOral GBC tab and Ceph/Pan GBC tab.
Corrected orientation problem for IO images in layout capture.
Version 1.0 Build 11
Release Date: 6/19/06
Made charges to commit image acquisition thus removing the problem of temporary images being maintained by the Gendex interface.
Version 1.0 Build 10
Release Date: 6/1/06
Modified how images are passed back and forth between XV and imaging extensions. Windows incompatibilities lead to a possible crash on some computers when passing images back and forth between the apps and extensions.
Version 1.0 Build 9
Release Date: 4/4/06
Added support for the Gendex eZ1 USB camera.
Added support for the Gendex FWT Fire wire camera.
Added additional GBC control tab so that now there is a separate GBC control for sensors/pan/ceph images.
Added hardware status window
Noise removal, intra oral GBC, pan/ceph GBC, flipping and inversion features added to the option menu.
Added Gendex device properties option for he user to configure Gendex devices through Gendex’s interfaces.
Version 1.0 Build 8
Release Date: -/-/06
Fixed a problem with DenOptix QST pan images where uses were getting the error “The image parameters could not be successfully retrieved from the Gendex image object. The image could not be captured”..
Version 1.0 Build 7
Release Date: 2/27/06
Corrected problem of missing images scanned from DenOptix.
Version 1.0 Build 6
Release Date: 1/4/06
Added gamma, brightness and contrast features.
Version 1.0 Build 5
Release Date: 11/15/05
Added auto-orientation of IO images when captured into a layout.