Added ability to view images full screen while in anatomical layout capture. (1659)
Added voice recognition support. (F1290)
Fixed issue with improper rotions during layout capture in anatomical view (B864)
Version 10 Build 17
Release Date: NA
Added some new imaging algorithms to available post capture filters. (F1390)
Version 10.0 Build 16
Release Date: 2/03/2012
Added support for Kodak Gip_SDK version 1.14.24. (F1312)
Fixed issue with images being labeled as retake with no tooth association during layout capture. (B716)
Version 10 Build 11
Release Date: 3/18/2011
Resolved issues with layout capture crashing, stop sensor button, and take/retake. (B460)
Resolved issue with setting proper image resolution. (B544)
Version 10 Build 9
Resolved image grabbing error and window focus issues. (B140)
Version 10 Build 8
Disabled the Done button while the sensor is active to avoid potential crashing. (B491)
Version 10 Build 7
Disabled the STOP button, since the new Kodak SDK does not support cancelling an acquisition. (B485)
Version 10 Build 5
Added support for Kodak SDK 184.108.40.206-A (F828)
Version 2.1 Build 4
Release Date: 12/2/09
Corrected a crash that could occur when rotating, inverting or flipping images during layout capture after the user clicked on the “Done” option.
Version 2.1 Build 2
Release Date: 9/14/09
Corrected an issue with pressing the <Enter> key on the keyboard when capturing a progression. The progression acquisition would end and return to the main application WITHOUT displaying the images.
Version 2.1 Build 1
Release Date: 8/21/09
Added new support for selected discipline in the invoking application.
Added new support for general anatomy selection, veterinary anatomy and veterinary dental fields.
Added CR modality support.
Ability to display patient names on the various Kodak extension interfaces (not all Kodak interfaces support displaying the patient name).
Version 2.0 Build 4
Release Date: 7/21/09
Updated with the latest version of the GUI controls.
Updated the image filter screen to delay the updates of the images while making changes to the filters. Added option to prevent auto-update all together. Added option to allow the user to update the image display manually.
Version 2.0 Build 37
Release Date: 3/3/09
Updated with the latest version of the GUI controls.
Corrected a memory leak that could potentially occur during progression acquisition.
Updated file naming problems where ‘-‘ in the computer names were being replaced with ‘_’. This interfered with the calibration image capture storage and retrieval.
Redesign of internal structure to remove conflict with the Kodak Image Filter being applied to images. Could result in an application crash during progression captures due to thread conflicts.
Corrected image rotation issue with continues endo series acqusitions.
Version 2.0 Build 35
Release Date: 8/22/08
Upgrader: removed the various SDK DLLs from being installed. The Kodak SDK DLLs should be installed via the installation of the hardware.
Upgrader: now adds the Kodak image filter for the DCV application.
Version 2.0 Build 35
Release Date: 8/12/08
Implemented new ghosting and retaken image classification. Additional options added to automatically mark images as ghosted or retaken based on the acquisition context.
Version 2.0 Build 34
Release Date: 4/8/08
Image Operations popup menu on the progression screen now has an “Advanced Image Display” option. This option is the same as double-clicking on an image.
Version 2.0 Build 33
Release Date: 3/26/08
Better image, series and study description labeling.
Added new option to quickly edit image notes during progression capture.
Version 2.0 Build 32
Release Date: 3/10/08
Dosimeter dialog added to appear after x-ray exposure. Dosimeter can be enabled and the length of time it appears can be changed in the IXDLL options.
RVG6000 button press option defaulted to off.
Version 2.0 Build 31
Release Date: 2/2/08
Updated the support of veterinary tooth display, tooth selection and information.
Version 2.0 Build 30
Release Date: 1/21/08
Corrected problem with first sensor initialization error being interpreted as a hardware error (which resulted in termination of the image acquisition). New version displays status messages when the first sensor initialization error messages are sent to the imaging extension.
Changes made to only display one message to the user when the connection to the hardware is lost.
Version 2.0 Build 29
Release Date: 1/9/08
Added new capabilities to display an advanced image display dialog when the user double-clicks on an image during progression capture.
Version 2.0 Build 27
Release Date: 9/6/07
Made changes to allow prompting for tooth information again when the user chooses to capture additional continuous endo progressions.
Version 2.0 Build 26
Release Date: 4/4/07
Corrected orientations when changing between various tiles when capturing a progression.
Version 2.0 Build 24
Release Date: 2/27/07
Image view control improved with additional functionality: image enhancement drop-down (auto BC, emboss, invert); drag gamma adjustment tool added.
Version 2.0 Build 23
Release Date: 1/8/07
Added popup that quickly allows the user to switch between multiple images that are associated with the same progression entry.
Added image operations to the progression capture screen (e.g. Invert, flip, rotate).
Display of a full screen image from the progression view now displays the image on a black background.
Display a different cursor when the user moves the mouse over the multiple indicator when a progression entry contains more than one entry.
Version 2.0 Build 22
Release Date: 12/5/06
Corrected values set for the Patient Orientation DICOM tag.
Version 2.0 Build 21
Release Date: 10/26/06
Added support for the new 6100 model sensor.
Version 2.0 Build 19
Release Date: 9/18/06
Added ghosted image support.
Version 2.0 Build 18
Release Date: 7/26/06
Added options to permit the user to modify the modalities that will be supported by the extension.
Version 2.0 Build 17
Release Date: 7/6/06
Modified the manner in which progression work when images are already captured for a tile. Old convention was to simply retake over existing images. The new convention skips over images that have already been taken when the user clicks on the “Capture Progression” button.
Version 2.0 Build 14
Release Date: 6/13/06
Made adjustments so that the image extension works with older generation sensors.
Version 2.0 Build 8
Release Date: 4/20/06
Added support for multiple shot selection progressions (i.e. the user can create a progression on the fly simply by clicking-and-dragging on a tooth chart).
Version 2.0 Build 7
Release Date: 4/19/06
If a progression tile does not have a sensor orientation associated with it, it is determined based on the Sensor Orientation Map.
Version 2.0 Build 6
Release Date: 3/8/06
New image viewer control integrated into extension.
Filter modalities now limited to extension’s modalities.
Full screen button graphic corrected.
Version 2.0 Build 5
Release Date: 2/3/06
16-bit image capture error corrected. In some circumstances an error is reported when converting 16-bit image data into a DICOM image object.
Spatial sharpening filter added to the post capture filter options.
Direct selection of progressions support added.
Version 1.0 Build 19
Release Date: 1/25/06
PA descriptions added to image, series and study notes.
Added reduction image filter option.
Version 1.0 Build 18
Release Date: 1/11/06
A new option was added to the hardware properties tab that will allow the user to defer sensor ready status until they press the button on RVG 6000 systems. This option does not affect use with the RVGui system. Note that if a RVG 6000 and RVGui system are plugged in at the same time then this option is ignored as well.
Version 1.0 Build 17
Release Date: 1/5/06
Added features to permit the progression name to be automatically added to the study comments, study description, series description and image comments tags.
Version 1.0 Build 16
Release Date: 12/15/05
Timeout error message corrected.
Stop progression capture option and done option now will terminate the capture.
Version 1.0 Build 14
Release Date: 11/16/05
Modified the program to use the sensor orientations in a progression instead of relying on the sensor orientation map during progression capture.
Version 1.0 Build 12
Release Date: 10/20/05
Corrected problem where disabled “Start Progression” button can be clicked again by user. This action could sometimes lead to a freeze of the application.
Version 1.0 Build 11
Release Date: 8/19/05
Added ability to click on numbered grouping on the progression view to select corresponding image.
Capabilities added to save images to the safety cache during a progression.
Miscellaneous changes and fixes.
Version 1.0 Build 9
Release Date: 7/26/05
Modified initial filters to correct for orientation issues with the images.
Version 1.0 Build 8
Release Date: 7/7/05
Correction made to prevent invalid UID generation.
Version 1.0 Build 7
Release Date: 7/5/05
Correction made for image resolutions. Some images had an improper resolution associated with an image.